; Topics: Advantest to Exhibit Wide Range of Semiconductor Test Solutions Enabling 5G Connectivity at SEMICON Japan on December. This is a Takeda TR (also known as Advantest R) programmable DC source, a great find at less than EUR 50, of course, in non. A quick look at Dave’s Advantest R Voltage and Current Source. Can it be used as a current reference for the uCurrent? A look at the.
|Country:||Moldova, Republic of|
|Published (Last):||16 April 2018|
|PDF File Size:||13.74 Mb|
|ePub File Size:||2.74 Mb|
|Price:||Free* [*Free Regsitration Required]|
Advantest R6142 Programmable DC Voltage/Current Generators
Turns out, one key is stuck! This is almost a advantes standard type of source, 0. Advantest Download or read online advantest r instruction manual. So, once back in Germany, we will just change the solder bridges in the voltage selector, and all will be ready for VAC mains.
It could be just dirt or age that gives low resistance. Looking inside, there is a logic board on the top side, and a shielded box with the analog circuits at the bottom. Advantest r instruction manual Advantest r instruction manual Link: I made a an offer for a Advantest R over a year ago on eBay.
Advantest R Programmable DC Voltage /current Generator | eBay
Download Advantest r instruction manual Information: Notify me of follow-up comments by email. As it turns out, the schematic is either incorrect or outdated, the coils on the transformer are VAC, and VAC.
This site uses Akismet to reduce spam. Advantest r instruction manual.
Advantest R – General Generators
Save my name, email, and website in this browser for the next time I comment. Leave a Reply Cancel reply Your email address will not be published. Unfortunately, these switches are difficult to disassemble, so I soaked the switch in some isopropanol.
The item arrived in good shape, but unfortunately, as described — non working, display comes up but no reaction to any key press. The keyboard is a matrix design. Seems the CPU is waiting for something — it is not executing any commands, when I look at the data bus with a scope probe.
Feb 3, – Forum: Learn how your comment data is processed. Read online or 6th edition manual mystery shopperSettings. All is built with really high precision metal encapsulated resistors, teflon standoffs, and DC filtering. A quick test — with a rr6142 instrument — it is working just fine also tested the current g6142.
Advantest R6144 Programmable DC Voltage /current Generator
This page is intended to be Is it an ebay score, or ebay junk? But what are calculations, if you can do measurements with the unit loaded. Your email address will not be published. The main reference is a LM, 0. A calculation shows the advaantest — it will definitely be better to configure it for Volts, rather than Volts, to avoid a dangerous over-voltage condition.
Notify me of new posts by email. What could it be? A quick look at the analog assembly to check for advantfst or other signs of aging — no dirt found.